"DIGITAL VHF / UHF SWR & POWER METER"
LIGHT WEIGHT, SLIM SIZE, HIGH RELIABILITY AND EASY TO OPERATE !!!
WIDE FREQUENCY RANGE / POWER RANGE!!!
EXCELLENT TEST EQUIPMENT FOR VHF & UHF RADIO MAINTENANCE SERVICE
. Direct / instant reading without calibration procedures in advance;
. Single button / simple operation;
. Back light / Data hold / Auto power off functions;
. Low insertion loss allows it to be connected permanently;
. Wide frequency range / Forward /Reverse power, SWR measurement that is applicable for the most of radio measurements;
. Frequency Range:
. VHF-1 (145 MHz ) / VHF-2 ( 220 MHZ) / UHF (440 MHz ) , cover ANYWHERE (100MHz to 500MHz)
We tested a TYT 220Mhz radio ( yes..we also sell them) with 63 Watts of power and a 1.3 SWR (See photos results below!))
. Power range: 0 to large 120 Watts RF ( Forward / Reflected Power ) in 2 different screens;
. Display: 3 1/2 LCD;
. SWR Range: 1.1 to 19.9;
. Insertion Loss: Less than 0.25dB;
. Impedance: 50 Ohms;
. Connector: N-female;
. Dimension: 67 x 69 x 37mm;
. Battery: 3 x AAA size battery ( included )
. Accessories: english instruction sheet
Specifications Dual-Band VHF / UHF / DIGITAL METER ..(NEW!!)
||VHF (145mhz) / UHF(435mhz)
COVERS FROM 100mhz~500mhz
||< 0.15dB(VHF) , < 0.25dB(UHF)
||0-70 degree C.
||3 x AAA Type Battery or Rechargeable <2mA(Backlight Off)
|Battery installation: Remove 4 black round screws on the corner of back cover and open it. Install 3 AAA batteries. BE CAUTIOUS to the polarities. Put the cover on and fasten the screws. IMPROPER INSTALLATION OF BACK COVER WILL CAUSE ERROR ON U-BAND MEASUREMENT!
(We include 3 x size AAA batteries)
Connect the Output of your transceiver to "TX" and Antenna or Load to "ANT". NOTE: UV-Band meter utilize "N" type connectors;
Proper adaptor/cable is required if your device has different type of connectors , we sell these @ $10 each if needed!
Press the red button to power on. After 2 seconds of LCD self test, it's ready for normal measurement.
UV-Band meter will display in sequence:"+ XXX P" (pass through power),"- XXX P" (feed back power)," XXX Γ" (SWR value), 2 seconds display for each mode. Press the red button will hold the current display. Press it again to continue the display.
Under the environment of night or weak illumination, just press and hold the red button for 1 second to turn the blue back light on. Press and hold the red button for 1 second again to turn it off.
8 minutes after no power was measured, it will power off automatically. Press and hold the red button for 2 seconds to turn off manually.
WE accept all major credit cards , including VISA , Mastercard and Discover with a 4% surcharge added to total.
Powered by Opencart
Peppers Radio Communication © 2017